In article the calculation example of expanded uncertainty in verification procedure of resistance thermo converters from Appendix GOST 8.461–2009 is considered. It is shown, that strict following to positions of this standard leads in an example to opposite result. The gross blunder reason – discrepancy «Guide to the expression of uncertainty in measurement» and its authentic transfer in the form of GOST R 54500.3–2011 positions of GOST R 50779.21–2004.
expanded uncertainty, confidence probability, confidence level
"Standartyzovannыi prymer rascheta neopredelennosty poverky termopreobrazovatelia soprotyvlenyia" [The standardized example of calculation of uncertainty verification of the thermo converter of resistance],
Information Processing Systems,